What should we pay attention to when using the laboratory chiller?
1. After receiving the goods, the FL +5°C ~ +35°C Water Chillers should first check whether the equipment has been hit during transportation, whether there are scars on the surface, and whether the display screen is intact.
2. After the arrival of the laboratory cooler, it is necessary to pay attention to whether its placement environment can be ventilated to avoid sunlight, and there is a certain space around the equipment on a stable cement floor.
3. Check whether the equipment is in normal condition after the laboratory cooler is started, and the operator needs to maintain and check the technology every year.
4. Before using the laboratory cooler equipment, read the operating instructions of the equipment to understand the function and name of each part of the equipment, and then operate it.
5. When selecting the heat conduction medium of the laboratory cooler, it is necessary to pay attention to whether its temperature range meets the working conditions of the project and whether it is suitable for equipment use. Generally, the manufacturer can be consulted to see whether the model is suitable.
6. There are many types of pipes for laboratory cooler equipment. Select which material is suitable according to the temperature requirements of the equipment model, and then select according to the length and diameter.
It is necessary to check the consumption of the pipes regularly, check whether there is damage, and replace them in time.
7. The heat conduction medium of the laboratory cooler needs to be replaced regularly after long-term use. It is recommended to use the same type of heat conduction medium to avoid the blockage of the laboratory heating and cooling machine caused by different types and different brands of heat conduction medium.
8. Before moving, repairing and maintaining the laboratory cooler, shut down the equipment and disconnect the power supply. Try not to tilt the equipment during moving.
The laboratory cooler is a fluid cooling and heating temperature control system, which can control the high and low temperature of materials.